|[AMRadio] GG 813 Linear in GE Ham News|
k4kyv at hotmail.com
Sun Oct 9 09:43:58 EDT 2005
>A standard carrier reading wattt meter (think 'Bird') should not move, when
>the carrier is modulated to 100%. A PEP reading wattmeter (it's my opinion
>that you can *not* find one that reads accuratly enough) willl only show
>that the rig -is- being modulated, but not if it is being over-modulated,
>causing distortion and splatter and being rather un-neighborly to others on
A field strength meter should not move when the carrier is modulated to
100%. It is designed to read AVERAGE carrier voltage. If the final stage
has good modulation linearity, the average carrier voltage doesn't change up
to 100% modualtion.
A true average-reading wattmeter will read increased output with modulation.
The power output includes carrier + sideband power. 1000 watts carrier
output should read 1500 (average, or mean) watts out with 100% sinewave
With a thermocouple rf ammeter in the line, the rf current at the output
should show a 22.5% increase with 100% sinusoidal modulation.
Many rf output meters actually read rf voltage output and therefore show no
upward kick with modualtion. An rf ammeter is a thermal device which does
not average the output, so it will show an upward kick with modulation.
Square the current increase to get the percent power increase.
More information about the AMRadio mailing list
This page last updated 20 Oct 2017.